TDS-3 Handheld Meter With Carrying Case
Brand: HM Digital
Dimensions: 6.1" x 1.25" x 1"
Weight: 0.5 lb
TDS-3 Handheld Meter With Carrying Case is a tool to measure Total Dissolved Solids (TDS) or the total amount of mobile charged ions, including minerals, salts or metals dissolved in a given volume of water. It is very portable, reliable and accurate in testing water and other solutions due to its advanced microprocessor technology. This handheld meter is ideal for all water purification applications, wastewater regulation, aquaculture, hydroponics, and colloidal silver. Perfect for both consumer or commercial use.
Check out a similar tester, TDS-EZ.* The U.S. EPA’s Maximum Contaminant Level of TDS for human consumption is 500 ppm.
TDS, which is based on conductivity, is expressed in parts per million (ppm) or milligrams per liter(mg/L). TDS includes any conductive inorganic element present other than the pure water molecules (H2O) and suspended solids. The higher the TDS level in water, the greater the probability of harmful contaminants that can pose health risks or hinder the absorption of water.
Key Features & Benefits:
* This is a typical TDS, Total Dissolved Solids Scale , what you can expect your water to be.
6.1 x 1.25 x 1 inches
0 - 9990 ppm (mg/L)
0 - 80° C
0.1° C(Celsius only)
0-999: 1 ppm; 1000-9990: 10 ppm (indicated by a 'x10' icon)
EC-to-TDS Conversion Factor:
NaCl (avg. 0.5)
Built-in sensor for Automatic Temperature Compensation of 1° to 50° Celsius(33° to 122° Fahrenheit)
2 x 1.5V button cell batteries (included) (LR44 or equivalent)
1000 hours of usage
Weight with case:
76.5g (2.7 oz)
Weight without case:
56.7g (2 oz)
- Manufacturer Warranty: 3 years
- HM Digital TDS3 meters come factory calibrated and are ready to use. They are designed to stay consistent. However, after prolonged usage, it may help to recalibrate your meter using a commercial standard NaCl-based solution, which is approximately, 0.5 µS of conductivity.
- Immerse the meter into the calibration solution. If the meter does not read within 2% of the calibration solution, adjust the reading by inserting a mini screwdriver (not included) into the trimmer pot (the hole on the back of the meter). Turn the trimmer clockwise to increase the reading and counterclockwise to decrease the reading. Note that the adjuster is very sensitive.
Tested and Certified